Flow rate indicator with large digits

A

U

P

Options

ZG - Coil input 5mVp-p

ZF - Coil input 10mVp-p

Accuracy: Resolution: 16 bit. Error < 0.01mA / ± 0.05% FS. Low level cut-off programmable.Span: 0.001 / 999,999 with variable decimal position.Update time: Four times per second.Voltage drop: Type A: max. 1V DC @ 20mA.Voltage drop: Type A - PL (loop powered): max. 2.6V DC @ 20mA.Relationship: Linear and square root calculation.Note: For signal type A and U: external power to sensor is required; e.g. type PD.

.

Power requirement

PL - Input loop powered from sensor signal 4 - 20mA (type A)

Type U: 0 - 10V DC. Analog input signal can be scaled to any desired range within 0 - 10V DC.Accuracy: Resolution: 16 bit. Error < 0.01mA / ± 0.05% FS. Low level cut-off programmable.Span: 0.001 / 999,999 with variable decimal position.Update time: Four times per second.Load impedance: Type U: 3kOhm.Relationship: Linear and square root calculation.Note: For signal type A and U: external power to sensor is required; e.g. type PD.

* Required field.